Press release

# High-Speed Recording of Frequency Trigger Points

09-20-2016 02:01 PM CET | Industry, Real Estate & Construction

## Press release from: CAS DataLoggers

ADwin Systems Log Data at Speeds of 25 Nanoseconds

At CAS DataLoggers we occasionally have customers in the power generation industry whose application involves a quickly-fluctuating waveform. These technicians need to accurately determine the frequency of a given waveform at each of its cycles. This can occur when manufacturers and servicers need to ensure the stability and reliability of their generators. For these applications, we offer real-time ADwin-Light Data Acquisition Systems for accurate frequency monitoring and analysis.

Identifying Fluctuating Waveform Frequencies:
In our most recent example, our customer is working in a power generation application and is investigating a rapidly-fluctuating waveform. This user wants to know its cause, so he needs a way to measure frequency easily. Here our given waveform takes the form of a sinewave, with peaks and valleys like those seen using an oscilloscope or when analyzing the AC power from a wall outlet.
Here our user is concerned with locating the waveform’s Trigger points, which mark the time between two given points in a waveform; for example two points immediately preceding each peak. In our application,

• Period is denoted as p
• Frequency is denoted as f
• The formula we use to find a given frequency is f = 1/p.

In this case, working with a power line’s frequency at 60Hz AC, the period of interest is equal to 16.667 milliseconds. This requires a high-speed system to record.

Capturing Trigger-Point Data:

Our customer is using the ADwin-Light-16 Real-Time DAQ and Control System for flexible signal generation. Consulting for this application, we first define a reference point called the trigger level. Our custom program does this by sampling the waveform very quickly (in this instance at 10 kZ).

The program commands the ADwin datalogger to start a clock, to identify the waveform’s trigger level, and then to sample data until it records the trigger point for a second time. In this way, the user then knows--at a high degree of accuracy--the time between those two points (the peak to peak). This also enables ADwin to calculate the frequency (f) of 1/p (here the trigger level).

ADwin acquires and evaluates input signals to monitor limit values. Users can also set up complex controls by using the ADwin’s capability of synchronously outputting multiple analog and digital signals on a time- or event-triggered base. The ADwin’s processor clock can capture data at extremely small increments as small as 25 nanoseconds. Depending on how fast we sample, we know the time length between the pair of events with extreme accuracy.

Frequency Monitoring Program:
Using the ADbasic programming environment, we developed a short program of just 40 lines which performs this frequency data acquisition. The program instructs the ADwin Light-16 to record the time duration between our two trigger levels. The program below took about 45 minutes to write:

DIM VAL,LAST_VAL AS LONG
DIM NEW_TIME, OLD_TIME, TIME_DIFF AS LONG
DIM TIME_DIFF_SUM, TIME_DIFF_AVG AS LONG
DIM WAIT_TIME AS LONG
DIM NUM_AVG_POINTS, AVG_COUNT AS LONG
DIM FREQ AS FLOAT
DIM TRIG_LEVEL AS LONG
DIM MIN, MAX AS LONG

INIT:
'Intialize values for sampling
GLOBALDELAY=4000 '10 kHz sample rate
TRIG_LEVEL = 35000 ' voltage (in A/D counts) for marking start/end of cycle
NUM_AVG_POINTS = 5 ' number of cycles to average
WAIT_TIME = 400000 ' Wait for 1/2 cycle(10 msec.)to prevent false trigger

'Intialize variable for averaging
TIME_DIFF_SUM = 0
AVG_COUNT = 0

EVENT:
IF (VAL > TRIG_LEVEL) THEN
TIME_DIFF = NEW_TIME - OLD_TIME
IF (TIME_DIFF > WAIT_TIME) THEN
PAR_1 = TIME_DIFF
OLD_TIME = NEW_TIME
TIME_DIFF_SUM = TIME_DIFF_SUM + TIME_DIFF
INC AVG_COUNT
IF (AVG_COUNT > NUM_AVG_POINTS)THEN
TIME_DIFF_AVG = TIME_DIFF_SUM / NUM_AVG_POINTS
TIME_DIFF_SUM = 0
AVG_COUNT = 1
ENDIF
FREQ = 1 / (TIME_DIFF_AVG * 25.0e-9)
FPAR_1 = FREQ
ENDIF
ENDIF

Benefits:

Many data acquisition systems on the market can sample frequencies, but ADwin hardware is capable of the high speed needed to capture all this frequency data in one cycle. Other data acquisition systems typically have to sample data over 5-10 cycles, having slow averages and too long a window to capture much data. Meanwhile ADwin’s processor clock and resolution can acquire frequency data at speeds of 25 nanoseconds, all during a single cycle for effective frequency analysis.

Additionally, ADlog software can log the values if required. ADwin has many post-measurement capabilities to work with the data—for example, users can view statistics such as standard deviation and Avg/Min/Max.

Computer Aided Solutions, LLC. dba CAS Data Loggers is a distributor of data loggers, paperless recorders and data acquisition equipment.

CAS DataLoggers, Inc.
8437 Mayfield Rd.
Chesterland, Ohio 44026
(440) 729-2570
(800) 956-4437
sales@dataloggerinc.com
www.dataloggerinc.com

This release was published on openPR.

Permanent link to this press release:

Please set a link in the press area of your homepage to this press release on openPR. openPR disclaims liability for any content contained in this release.

### You can edit or delete your press release High-Speed Recording of Frequency Trigger Points here

News-ID: 364690 • Views:

## More Releases from CAS DataLoggers

Ensuring Workplace Safety: Data Loggers for Compliance With California Regulatio …
In workplaces across California, ensuring the health and safety of employees is paramount. This commitment is not just a moral imperative, but a legal requirement under California Code of Regulations Section 3395, which mandates specific measures to ensure workplace safety by protecting workers from heat illness. Among these measures is the monitoring of environmental conditions such as temperature and relative humidity, critical factors that can significantly impact employee well-being. Understanding California
New AirGate 4G Cellular Router from Novus
NOVUS presents AirGate 4G, an industrial VPN router for cellular networks. Data sending is secure with this new device as it uses encryption protocols and firewall systems most commonly used in IT infrastructures, including automatic fallback for 4G, 3G, and 2G cellular networks. AirGate 4G is CE Mark certified and was developed for industrial environments. It can maintain its high availability performance even in extended operation situations, being equipment suitable
New AERINOS ADS-300 from Infinite Informatics
New Wireless NB-IoT/LTE-M Data Collection End Node CAS DataLoggers is pleased to introduce the new AERINOS ADS-300 wireless sensor end node from Infinite Informatics. The ADS-300 is designed to transmit data via either NB-IoT/LTE-M cellular networks and is targeted at remote monitoring applications. NB-IoT is a wireless protocol designed for the Internet of Things that offers very low-power operation using either existing GSM or newer LTE cellular technology. NB-IoT is a
New TR32B Bluetooth Data Logger From TandD
CAS DataLoggers is pleased to announce the updated TR32B log-EZ as an effective low-cost solution for temperature and humidity measurement. The log-EZ is designed as a compact, user-friendly data logger to easily measure and record ambient temperature and humidity. It provides Bluetooth communication and a smartphone/tablet app that can display and upload data to the cloud. The TR32B provides a measurement range of 0° to 50° Celsius (32° to 122°

All 4 Releases