Press release

# Thermistor Calibration for High Accuracy Measurements

07-25-2018 11:10 AM CET | Industry, Real Estate & Construction

## Press release from: CAS DataLoggers

Calculating Steinhart-Hart Coefficients for High-Precision Readings

Technicians and engineers often use thermistors to measure temperature in applications which require high accuracy. Thermistors operate by changing resistance as their temperature changes in a very predictable but non-linear way. This characteristic allows them to provide higher accuracy than thermocouples or RTD’s. In order to ensure this high accuracy, thermistor calibration is an important consideration.

One challenge when using thermistors is calculating the temperature from the measured resistance value. To accomplish this, the Steinhart–Hart equation is used to convert a thermistor sensor’s resistance to temperature. When compared against other methods, Steinhart-Hart models will give you much more precise readings across the sensors’ temperature ranges, often within a few hundredths of a degree. Although the Steinhart-Hart equation is not universally known, it is useful in data logging applications such as measuring lake water temperatures, solar hot water systems, and skin temperature measurement.

Many high quality data loggers such as the dataTaker DT8x, Grant SQ20xx and VersaLog VL-TH allow you to enter the coefficients to automatically derive temperature from measured thermistor resistance. As part of our free tech support, we at CAS DataLoggers often provide help in this area for customers who call in asking how to perform the conversion.

Thermistor manufacturers don’t always provide users with Steinhart–Hart coefficients for their sensors; they may simply provide resistance versus temperature tables. In the case of a manufacturer-provided table, it’s not immediately obvious how to derive the necessary coefficients. Or, the user may want to perform self-validation of thermistors by measuring the resistance at several known temperature points and use this data to derive the Steinhart-hart coefficients. To speed up the process, there are several Steinhart-Hart calculators online which allow you to enter the temperature and resistance values and then generate the coefficients. You’ll find a link to our own online calculator, along with an example table, at the end of this article.

Deriving Steinhart-Hart Coefficients for Thermistor Calibration:
In cases where the Steinhart–Hart coefficients are not provided by your thermistor manufacturer or if you are doing thermistor calibration, you can derive them yourself. First, you’ll need three accurate resistance values (either from a table or measured) at three known temperatures and then insert them into the formula to derive the A, B and C coefficients.
Our engineers developed a Steinhart-hart thermistor coefficient calculator in the form of an Excel spreadsheet that does this automatically. The figure below shows an example of how to enter your actual temperature in Celsius and Kelvin (Col. B and C) and your sensor’s resistance (Col. D) at 3 reference points (T1-T3) to derive the 3 Steinhart-Hart coefficients (A, B, C).

You can obtain a copy of this spreadsheet by clicking here.

To get more information on thermistor calibration or thermistor data loggers, contact a CAS Data Logger Applications Specialist at (800) 956-4437 or visit the website at www.DataLoggerInc.com.

Computer Aided Solutions, LLC. dba CAS DataLoggers is a distributor of data loggers, paperless recorders and data acquisition equipment.

We have the industry’s most complete selection of data logging equipment, with hundreds of different models from more than 18 manufacturers. With data loggers from 1 to 300 channels we can record temperature, humidity, force/strain, pressure, flow, voltage, current, resistance, vibration and other digital signals, in connection with serial (RS-232/RS-485), CAN/OBD or SDI-12 devices. We sell directly to end users and also work through a network of distributors and resellers throughout the United States, Canada, Central and South America.

CAS DataLoggers
8437 Mayfield Rd Unit 104
Chesterland, OH 44026

This release was published on openPR.

Permanent link to this press release:

Please set a link in the press area of your homepage to this press release on openPR. openPR disclaims liability for any content contained in this release.

### You can edit or delete your press release Thermistor Calibration for High Accuracy Measurements here

News-ID: 1143817 • Views:

## More Releases from CAS DataLoggers

New AirGate 4G Cellular Router from Novus
NOVUS presents AirGate 4G, an industrial VPN router for cellular networks. Data sending is secure with this new device as it uses encryption protocols and firewall systems most commonly used in IT infrastructures, including automatic fallback for 4G, 3G, and 2G cellular networks. AirGate 4G is CE Mark certified and was developed for industrial environments. It can maintain its high availability performance even in extended operation situations, being equipment suitable
New AERINOS ADS-300 from Infinite Informatics
New Wireless NB-IoT/LTE-M Data Collection End Node CAS DataLoggers is pleased to introduce the new AERINOS ADS-300 wireless sensor end node from Infinite Informatics. The ADS-300 is designed to transmit data via either NB-IoT/LTE-M cellular networks and is targeted at remote monitoring applications. NB-IoT is a wireless protocol designed for the Internet of Things that offers very low-power operation using either existing GSM or newer LTE cellular technology. NB-IoT is a
New TR32B Bluetooth Data Logger From TandD
CAS DataLoggers is pleased to announce the updated TR32B log-EZ as an effective low-cost solution for temperature and humidity measurement. The log-EZ is designed as a compact, user-friendly data logger to easily measure and record ambient temperature and humidity. It provides Bluetooth communication and a smartphone/tablet app that can display and upload data to the cloud. The TR32B provides a measurement range of 0° to 50° Celsius (32° to 122°
New A2-06 Temperature & Alarming Monitoring Pod by CAS
Medical storage applications depend heavily on monitoring the temperature of their goods in cold storage to protect items such as vaccines, blood, medicine, and tissue samples. CAS DataLoggers is pleased to introduce the A2-06 Ethernet Wired Temperature Measurement Pod designed for measuring temperatures in medical refrigerators, freezers, incubators, and in cryogenic storage. The pod is an Ethernet LAN-wired temperature monitor with connections for two external RTDs and a thermocouple sensor.

All 4 Releases

## More Releases for Data

HOW TO TRANSFORM BIG DATA TO SMART DATA USING DATA ENGINEERING?
We are at the cross-roads of a universe that is composed of actors, entities and use-cases; along with the associated data relationships across zillions of business scenarios. Organizations must derive the most out of data, and modern AI platforms can help businesses in this direction. These help ideally turn Big Data into plug-and-play pieces of information that are being widely known as Smart Data. Specialized components backed up by AI and
Global Data Analytics Outsourcing Market |data analytics outsourcing, big data o …
Market Research Reports Search Engine (MRRSE) has been serving as an active source to cater intelligent research report to enlighten both readers and investors. This research study titled “Global Data Analytics Outsourcing Market “ The report on data analytics outsourcing market provides analysis for the period 2016 – 2026, wherein 2018 to 2026 is the forecast period and 2017 is the base year. The report covers major trends and technologies playing
Test Data Management (TDM) Market - test data profiling, test data planning, tes …
The report categorizes the global Test Data Management (TDM) market by top players/brands, region, type, end user, market status, competition landscape, market share, growth rate, future trends, market drivers, opportunities and challenges, sales channels and distributors. This report studies the global market size of Test Data Management (TDM) in key regions like North America, Europe, Asia Pacific, Central & South America and Middle East & Africa, focuses on the consumption
Data Prep Market Report 2018: Segmentation by Platform (Self-Service Data Prep, …
Global Data Prep market research report provides company profile for Alteryx, Inc. (U.S.), Informatica (U.S.), International Business Corporation (U.S.), TIBCO Software, Inc. (U.S.), Microsoft Corporation (U.S.), SAS Institute (U.S.), Datawatch Corporation (U.S.), Tableau Software, Inc. (U.S.) and Others. This market study includes data about consumer perspective, comprehensive analysis, statistics, market share, company performances (Stocks), historical analysis 2012 to 2017, market forecast 2018 to 2025 in terms of volume, revenue, YOY
Long Term Data Retention Solutions Market - The Increasing Demand For Big Data W …
Data retention is a technique to store the database of the organization for the future. An organization may retain data for several different reasons. One of the reasons is to act in accordance with state and federal regulations, i.e. information that may be considered old or irrelevant for internal use may need to be retained to comply with the laws of a particular jurisdiction or industry. Another reason is to
Data Quality and Data Governance Solution Market - Demand For Cost-Effective Dat …
In the enterprise data management ecosystem, data quality is a broad term which refers to the quality, integrity, and consistency of data and/or process etc. Data quality also implies the degree of data accuracy and consistency. On the other hand, data governance focusses on the management of data assets by assigning authority, control, and responsibility of data and encompasses three key areas: people, process, and technology. Data quality and data governance